Savia Art - Frequently Asked Questions
Information regarding the technical aspects of Phonak hearing aids relevant to Audiology and the hearing aid fitting.
- Easy FM
- 5 manual programs
- DataLogging with Self Learning
- High Resolution Noise Canceler
- Advanced Feedback Phase Inverter (NoWhistle Technology)
- Wind Noise Supressor
- Sound Relax
- Audiogram Direct
- Microphone modes: Digital SurroundZoom, Real Ear Sound, Fixed Directional, Omni Directional
Savia Art has 4 automatic programs in the AutoPilot (calm, speech in noise, comfort in noise, and music). There are also 2 Easy Programs (Easy Phone and Easy FM). In addition, there are 5 additional manual programs that can be accessed through the tactronic switch or the remote control.
The end of battery life beep cannot be turned off. Digital Bionics products end-of-life battery life process is as follows:
Once a low voltage level is reached, an acknowledgement beep is heard and DataLogging is turned off. This beep is, in fact, 2 beeps separated by 2 seconds. The hearing instrument is muted between the two beeps. As battery current falls further, every 30 minutes a beep will be heard, reminding the wearer to change the battery. If the battery is not changed and the battery current reaches a critical level, one more end-of-life beep is heard, all manual controls are disabled and the power-down process is started. This process involves a decrease in gain of 5 dB, every 10 seconds. After this, if the battery has not been changed, the instrument will turn off.
The Savia Art BTE (except micro) products can be locked, ensuring that the battery cannot be inadvertently removed from the device. The hearing instrument will not be able to be turned off using the battery door once the battery door is locked, turn the screw to the horizontal position to be able to turn the aid off. The child lock system uses the same opening as the right/left indicator. The child safety screw has to be mounted by Phonak.
The Digital Bionic products offer a range of acoustic acknowledges, or beeps. These can be changed in terms of frequency and loudness. Within the schematic, acoustic acknowledges are inserted before the gain model, therefore the beep loudness is hearing loss dependent. Limiting (MPO) is applied to and affects the loudness of the beeps, too. Turning the beeps off applies only to manual interventions, not to the AutoPilot melody, warning beeps such as end-of-life, different input source (EasyPhone, EasyFM) and max/min volume reached.
- AutoPilot: melody
- P1: 1 beep
- P2: 2 beeps
- P3: 3 beeps
- P4: t-coil + mic: low-high-high (not accessible with KeyPilot)
- P5: t-coil: low-high (not accessible with KeyPilot)
- VC 1 step up or down (BTEs): 1 beep
- VC preset MCL reached: 1 beep at a lower frequency
- VC maximum range: 2 beeps
Savia Art BTE’s (except micro) are compatible with the ML9i receiver. The AS9 audioshoe can also be used with a universal receiver. The MyLink will also work with all Savia models with a t-coil. The battery door will need to be changed to the FM battery door for use with the audioshoe or receiver.
Yes, the Savia Art BTEs (except micro) can be used with either a wired Cros/BiCros or a wireless CrosLink System. For the wired Cros, the components are the hearing instrument, audioshoe, Cros cord, and offside mic. For the CrosLink, the components are the hearing instrument, audioshoe, CrosLink Transmitter and CrosLink Receiver.